CONTENTS. Service Manual ECG C Contents The CE mark is applied only to the ECGL/M/N Electrocardiograph. This equipment complies. Service Manual ECG Composition. · To order a replacement assembly above, use the Code No. · To order a replacement component inside an. View Homework Help – Nihon_Koden_Cardiofax_ECG_-_Repair_manual from BA at Vietnam National University, Ho Chi Minh City. ECGL.
|Published (Last):||7 February 2013|
|PDF File Size:||12.27 Mb|
|ePub File Size:||11.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the solvent or toxic substance touches your skin or gets into your eye or mouth, immediately 9620 it with a lot of water and see a physician.
Use a piece of cotton moistened with alcohol to clean the motor rotation sensor. The results of the analysis may not correspond to the judgment of a physician.
Battery 12V 2Ah for ECG CARDIOFAX 1250-9620-9629 NIHON KOHDEN
Motor rotation sensor is clean. Faulty ECG control board. The instrument does not operate during battery power operation.
The fuse rating is shown on the fcg. Most of the boards are multilayer boards with surface mount electrical devices, so the mounting density of the board is too high.
Enter the system test number  Test level 1 or  Test 960l 2. No No Set the recording paper. Removing the Magazine and Recording Paper Otherwise, the operator may get an electrical shock.
Nihon Kohden ECG – ECG
Heart rate display is correct. The Test level 2 is called up and the instrument is in standby mode for entering the system test number. Faulty cable connection Replace the LCD unit. Remove the five M3 binding head screws.
After all LEDs light up, they go out one at a time. Measuring and Test Equipment Maintain the accuracy of the measuring and test equipment by checking and calibrating it according to the check and calibration procedures. The power board is faulty. Do not connect the battery pack to an AC outlet or lighter socket in a car.
Power off time out. Instrument does not affect surrounding fcg. Removing the Thermal Head Repeat steps 2 to 4 to enter the new numbers for the other settings. Replace the power fuse with a new one. NOTE The following settings are 9620, reset to the factory initial settings even if the instrument is initialized.
Removing the Thermal Head and Motor Assy Battery pack is in the instrument. The instrument does not operate on AC power. Adjust the thermal head position.
10 Lead EKG Cable Connector Fit Nihon Kohden Cardiofax Banana 4 15pins Connector
The battery pack short-circuits, the substance liquid comes out and the battery pack explodes. To reduce this unwanted radiofrequency signal, attach the optional ferrite core to the cable of the external instrument. Liquid ingress The instrument is not waterproof, so do not install the instrument where water or liquid can get into or fall on the instrument.
Replace the ECG control board. After the test, the instrument returns to the standby mode for entering the system test number.