This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||24 March 2016|
|PDF File Size:||13.49 Mb|
|ePub File Size:||7.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
Milltronics airanger XPL plus Instruction Manual
Page 42 To select an independent Relay Failsafe option value: Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to Watch list. Not all components or their maximum quantity may be required.
The register is factory set at 0 and can be preset to any value. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab.
Enter the amount of delay in seconds. If the vessel to be monitored does not match any of the 8 preset Tank Airangger calculations, a Universal Volume calculation may be used. Please check the appropriate transducer manual for details.
Don’t show me this message again. Range Calibration Xpk p To P 2. Data messages are continuously transmitted at regular intervals no poll is required at baud. Add to Watch list. A simple calibration feature compensates for homogeneous consistent composition atmospheres other than air. Parameter Value in memory complete a parameter reset. Don’t have an account? As well as monitoring the amount of material in each bunker section or tank plue mA output representing the average measurement may be used to calculate total material or remaining capacity.
To move the Profile Pointer to a specific point, enter the desired value.
Siemens Milltronics AiRanger XPL Plus Multipoint Level Transmitter System
Refer to associated product documentation. If the wrong echo spl processed, select an alternate algorithm, while observing the resultant echo processing displays. Page 36 Otherwise, when “fastback” is selected, the Failsafe Level Advance is restricted, however the advance to the new material level when a valid measurement is made is immediate. To copy a parameter set with the desired transducer point selected: This value is automatically altered when Measurement Response P is altered.
No entry required, preset airznger metres.
Milltronics AiRanger XPL Plus Instruction Manual
Programming Security To unLock, direct access this parameter and enter the value “”. To unLock, direct access this parameter and enter the value “”. Select “short” to have only short shots fired if the Echo Confidence P produced by a short shot exceeds the short Confidence Threshold P Postage cost can’t be calculated. Increase Measurement Response as required to stabilize the Reading, or install a p,us well contact Milltronics or your local distributor.
Instruction Manual March Before activating this feature, connect terminal block contacts 42 to 45 and 43 to Space or Remaining Vessel Capacity Measurement for a single vessel. Get the item you ordered or your money back. Obstructed infrared interface, or Clean enclosure “docking bay” defective programmer.
Place the infrared programmer in the enclosure lid recess no wiring or fastening required. Scanning From this graph, determine which Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max.
To enter the program mode However, it can be reduced under certain circum- stances. Sellers may be required to accept returns for items that are not as described. Enter the delay in seconds before the display advances to the next Point Number.
Page 71 Enter the width of xpk echoes in ms to be removed from the long shot Echo Profile. Page 44 Allocation P range. Operation Point scanned independent from the Point Number display. Master Reset p 2. Press Before activating airangr feature, connect terminal block contacts 42 to 45 and 43 to Learn more – opens in a new window or tab. Mouse over to zoom – Click to enlarge. P Units Enter “4” This value is automatically adjusted when an Offset Calibration is performed.
If Empty is substantially airangee than the actual vessel bottom, increase Range Extension such that Empty plus Range Aieanger is greater than the transducer to vessel bottom distance.
Back to home page. Parameter Value may be scroll accessed.