The CDB is a dual precision monostable multivibrator with independent trigger and reset controls The device is retriggerable and resettable and the control. B Monostable Multivibrator are available at Mouser Electronics. Mouser offers inventory, pricing, & datasheets for B Monostable Multivibrator. Data sheet acquired from Harris Semiconductor. SCHSE. Features. • Retriggerable/Resettable Capability. • Trigger and Reset Propagation Delays.
|Published (Last):||17 May 2010|
|PDF File Size:||14.21 Mb|
|ePub File Size:||11.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
MC14538B Datasheet PDF
Message 3 of Thank you for uploading the circuit, I have opened and compared it with the ones I found online for the CD, and they look pretty much the same, so should I assume that you have already looked at the datasheets of both components to determine if the pinout is the same?
Message 10 of Message 4 of Message 2 of Message 1 of Designed to simplify your testing by covering most of the different needs of a modern test lab with one sensor.
B Datasheet, PDF – Alldatasheet
Message 5 of Read more about cookies. Auto-suggest helps you quickly narrow datashest your search results by suggesting possible matches as you type.
Most Active Software Boards: Get Waves directly in your inbox. Message 9 of So this is probably use less at this point, but in the most recent version of Multisim Yeah datadheet having trouble connecting CD to use it as CD as shown in the picture.
Subscribe to Waves and get the stories from the world of sound and vibration Get Waves directly in your inbox Industry innovation Customer case studies. TEDS function and the possibility of mounting with an M3 stud speed up test setup. I cannot see the attachment. Can you elaborate datqsheet little bit more on your post? Message 8 fatasheet It seems the pinout layout for the CD in the multisim is quiet different than the actual CD If you found that they can be used as a sustitute I would think that there schematic should be very similar to get the response that you want, but the confirmation of the wiring is somehting that you can get from the devices datasheet.
This is the diagram that I was working on. I am glad to see that you found the documets helpful! When only a single- or bi-axis measurement is needed at a location, the small size and possibility of single axis powering supply makes Type B the right choice. So were you able to find the workaround afterwards?
Subscribe to Waves and get the stories from the world of sound and vibration. datashert
multisim using CD – Discussion Forums – National Instruments
CD was used because it was comparable with CDs. Are you trying to connect the CD using the diagram for the CD?
Would you be able to show me how to connect it? The wide frequency range, from 0.
Message 6 of I am not sure how to connect each other.